SSCOR Duet with Retention Bracket
AC suction unit
Battery back up
The retention bracket feature allows the Duet to be placed in a convenient location on the crash cart and save valuable work space
When fully charged and at full capacity the battery delivers approximately 45 minutes (+/- 10%) running time
>30 lpm clinical airflow
Vacuum range <50mmHg to >525mmHg
Meets reduced maximum recommendations of 80-120 mmHg vacuum for tracheal and pediatric suction prodedures
Vacuum as low as <50mmHg for neonates
Dual mode battery charger activates when plugged into AC power source
Battery charger drops to float when battery is fully charged
Clearly labeled battery level indicator
1 year warranty, 3 year battery replacement policy
Lightweight: 10.65 lbs/4.83 kgs
Large carrying handle
Can be customized to fit your canister
Sealed lead acid battery making battery replacement affordable
Performance: >30 lpm clinical airflow, <50 mmHg to > 525 mmHg vacuum
Dimensions: 9" H x 5.25 W x 17 L (22.9 cm H x 13.3cm W x 43.2cmL)
Weight: 10.65 lbs/4.83 kgs Device only
Charger: Internal dual mode battery charger activates when connected to AC power, dropping back to float when battery reaches full capacity. LED's indicate charging status and remaining running time.
Power: AC: 100 – 240V AC, 50 – 60Hz DC: 12V DC Sealed Lead Acid Battery
Battery: 12v DC Sealed Lead Acid Battery
Run Time: 45 minutes +/- 10% when battery fully charged at full capacity
Input: 100-240V AC, 1.0-0.5A, 50-60 Hz
Output: 40 watts 12V DC 3.3A
STANDARD ITEMS INCLUDED WITH PRODUCT: Retention bracket, canister, 2m patient tube, HI-D suction tip, AC power cord and Quickstart Guide.
CONFORMS TO: ETL listed and conforms to ISO 10079-1, UL 60601-1, IEC 60601-1, and EN 60601-1-2 and certified to CAN/CSA STD C22.2 NO. 601.1-M.90, IPX4 per IEC 60529, EN1789:1999.
REFERENCE / MODEL NUMBER: 2314B
SSCOR suction units and components containing sealed lead acid batteries contain lead. These products can expose you to Lead which is known to the State of California to cause cancer. For more information, go to www.P65Warnings.ca.gov.