SSCOR DCell Suction
Hours of Medical Vacuum Using D Cell Alkaline Batteries
Designed for disaster preparedness, the DCell runs for six to nine hours on a set of off the shelf alkaline D-Cell batteries. A 60 second battery change gives you another six to nine hours of quality medical vacuum.
Variable regulator and gauge
Battery condition indicator
Does not need to be kept on charge
Power options include: 12V DC power using the SSCOR #80665 DC power cord or AC power using the SSCOR #200-00001 AC power converter.
Accommodates most standard hospital suction canisters
Canister can be oriented in one of three positions
3 year warranty
Performance: Clinical airflow >=30 lpm, <50 mmHg to >525 mmHg vacuum (specifications based on tests performed with fresh Duracell alkaline batteries)
Dimensions: 12”H x 9”W x 11”D (30.48cm H x 22.86cm W x 27.94 cm D)
Vacuum Pump: 12V DC; Exceeds -525mmHg for the first few hours and exceeds 300mmHg for the stated run time*
Weight: 9.5 lbs/4.3 Kg (when batteries are installed; batteries are not included)
External Power Source Adapters: Power options include: 12V DC power using the SSCOR #80665 DC power cord or AC power using the SSCOR #200-00001 AC power converter.
Power: 12V DC 1A
Battery: Off- the-shelf 12xD Cell Alkaline Batteries (not included)
Run Time: 6-9 Hours*
Standard Items Included With Product: 1200cc Canister, 2m patient tube, HI-D® Suction Tip and Quickstart Guide
Conforms To: EC 60601-1, IEC 60601-1-2, IPX4, ETL Listed
Reference / Model Number: DM10-001
Batteries not included.
*Run time is defined as the period of time that the device generates more than 300 mmHg negative pressure. Total run time typically varies between 6-9 hours. These run time specifications are based upon tests using new Duracell® CopperTop batteries. Run time will vary depending on battery manufacturer, state of the charge, ambient temperature, and age of the batteries.
SSCOR warrants that each new product is free from defects in material and workmanship under normal use and service for a period of three years from the date of purchase. If returned to SSCOR, we will arrange for repairs or replacement within the terms of the warranty. The product should be decontaminated and returned properly packaged and postage prepaid. Loss or damage in transit to the factory shall be at the purchaser’s risk. Please call 800-434-5211 or international +1 818-504-4054 for return authorization or for the location of an authorized repair center. Loss or damage in return shipment from SSCOR shall be at the purchaser's risk.
The warranty shall not apply to any SSCOR product which has been repaired by anyone other than an authorized SSCOR representative, or altered in any way so as, in SSCOR’s judgment, to affect its safety or efficacy, nor which has been subject to misuse, negligence, or accident, nor which has had the serial number altered, effaced or removed.
Neither shall this warranty apply to any SSCOR product which has been connected otherwise than in accordance with the instructions furnished by SSCOR.
This warranty is in lieu of all other warranties expressed or implied and of all other obligations or liabilities on SSCOR’s part, and SSCOR neither assumes, nor authorizes any representative or other persons to assume for it, any other liability in connection with the sale of SSCOR products.
This warranty gives you specific legal rights and you may also have other rights that vary from jurisdiction to jurisdiction. For countries where minimum warranty terms are determined by statue, the warranty term is the longer of the statutory period or the term listed above.
All disposable items including collection canisters, patient tubing and catheters are excluded from this warranty.