SSCOR DCell Suction
Hours of Medical Vacuum Using D Cell Alkaline Batteries
Designed for disaster preparedness, the DCell runs for six to nine hours on a set of off the shelf alkaline D-Cell batteries. A 60 second battery change gives you another six to nine hours of quality medical vacuum.
Variable regulator and gauge
Battery condition indicator
Does not need to be kept on charge
Power options include: 12V DC power using the SSCOR #80665 DC power cord or AC power using the SSCOR #200-00001 AC power converter.
Accommodates most standard hospital suction canisters
Canister can be oriented in one of three positions
3 year warranty
Performance: Clinical airflow >=30 lpm, <50 mmHg to >525 mmHg vacuum (specifications based on tests performed with fresh Duracell alkaline batteries)
Dimensions: 12”H x 9”W x 11”D (30.48cm H x 22.86cm W x 27.94 cm D)
Vacuum Pump: 12V DC; Exceeds -525mmHg for the first few hours and exceeds 300mmHg for the stated run time*
Weight: 9.5 lbs/4.3 Kg (when batteries are installed; batteries are not included)
External Power Source Adapters: Power options include: 12V DC power using the SSCOR #80665 DC power cord or AC power using the SSCOR #200-00001 AC power converter.
Power: 12V DC 1A
Battery: Off- the-shelf 12xD Cell Alkaline Batteries (not included)
Run Time: 6-9 Hours*
Standard Items Included With Product: 1200cc Canister, 2m patient tube, HI-D® Suction Tip and Quickstart Guide
Conforms To: EC 60601-1, IEC 60601-1-2, IPX4, ETL Listed
Reference / Model Number: DM10-001
Batteries not included.
*Run time is defined as the period of time that the device generates more than 300 mmHg negative pressure. Total run time typically varies between 6-9 hours. These run time specifications are based upon tests using new Duracell® CopperTop batteries. Run time will vary depending on battery manufacturer, state of the charge, ambient temperature, and age of the batteries.